Acoustic analysis of MRI scanners
Keywords:Acoustic impedance, Boundary conditions, Boundary element method, Ducts, Flanges, Frequencies, Magnetic resonance imaging, Mathematical models, Pressure effects, Resonance, Scanning, Velocity measurement, Wave propagation, Acoustic pressure, Acoustic resonances, Sound pressure, Velocity amplitudes
AbstractThe acoustic noise radiation from gradient coils in MRI scanners was investigated using a analytical model of finite cylindrical ducts with infinite flanges. The radiation impedances was calculated for a finite cylindrical duct with rigid and absorptive walls. The total sound pressure inside the duct can be calculated by satisfying the boundary conditions at the wall and the open ends. The five peaks in frequency spectra was observed due to wave reflections at the open ends that reach their maximum value at the cut-off frequency and causes acoustic resonance.
How to Cite
Copyright on articles is held by the author(s). The corresponding author has the right to grant on behalf of all authors and does grant on behalf of all authors, a worldwide exclusive licence (or non-exclusive license for government employees) to the Publishers and its licensees in perpetuity, in all forms, formats and media (whether known now or created in the future)
i) to publish, reproduce, distribute, display and store the Contribution;
ii) to translate the Contribution into other languages, create adaptations, reprints, include within collections and create summaries, extracts and/or, abstracts of the Contribution;
iii) to exploit all subsidiary rights in the Contribution,
iv) to provide the inclusion of electronic links from the Contribution to third party material where-ever it may be located;
v) to licence any third party to do any or all of the above.